Constant temperature water bath with exocyclic coolnit bath
Thermo supplier EZ-101
Code No.： 0043201-000
External circulation while using water bath is available. Addressed to high temperature, silicon oil can be used as well. EZ-101 for high temperature range.
Applications●Circulation to device which need temperature regulation at high temperature.
●Quality test at high temperature using water bath.
●Moisturizing and Cooling for samples.
Features●Silicon oil can be used in the heat insulated water bath.
●External circulation function by variable flow rate. Wetted parts are made of stainless steel overall.
●Available from -30℃ – +200℃ by combining the optional immersion cooler “Cool pipe” (when setting +30℃ or more, cool pipe is removed).
|Temperature range||+60℃ to +200℃ (*1) (*2)|
|Control accuracy||±0.1 - 0.3℃ (*3)|
|Pumping capacity(50/60Hz)||Max.flow rate: approx. 8/10 L/min (variable)|
Max. lift height: approx. 4/5 m
|Nozzle diameter||Outer diameter φ10mm|
|Safety device/function||Short/Over current breaker, Fuse (3A), Floating type boil without water protection (with protection of chattering), Thermosense type boil without water protection, Thermal switch for motor protection, High and Low temperature protection of sample, Diagnosis of temperature sensor error (disconnection/short), Diagnosis of low water level error, Alarm for high and low temperature protection of sample|
|Other Functions||Agitating inside water bath, Cooling pipe (Outer diameter φ8mm) (*2), Immersion cooler port|
|Dimensions inside bath||170×300×145Hmm|
|Effective dimensions inside bath||150×100×140Hmm|
|Water bath capacity (water level 80%)||approx. 6L|
|Power Supply||AC100V, 13A|
|Standard Accessory||Stainless steel lid×1|
|(*1)Temperature setting is available between -30ºC - +200 ºC. When setting temperature lower than RT, please use Immersion cooler together. There is special opening for Immersion cooler. When setting below +7ºC, please use antifreeze. When setting above +70ºC, please use heating medium for high temperature.|
|(*2)When setting around RT - +60ºC, please circulate cooling water (tap water) in cooling pipe. Temperature rise by work heat from mixing pump need to be minimized at below +60ºC.|
|(*3)Performance may not be maintained due to environmental temperature, power supply etc.|
|•Please do not use pure water or distilled water. These may cause breakdown of machine.|