Constant temperature Chambers (Small size)
In vitro Box iB-130
Code No.： 0077960-000
High quality and Energy saving Refrigerator Incubator. Flagship Small/Middle-size Constant temp. chambers. Possible for operation of small devices inside the chamber.
Applications●Storage of Samples and Reagents
●Cultivation and Immunoprecipitation by placing on a Shaker or Rotator
Down Load（Useful Information）General Catalog
High quality In vitro Box unique to TAITECInside and outside are made of stainless steel. It ’s highly durable and easy to clean.
Small size shaker and culture rotator can be entered.Small size shaker(NR-1/2/3 etc.) can be placed at the bottom of iB-130. Only devices with low vibration（RT series etc.）can be placed on the upper stage. There is one outlet (up to 3A) inside.
Functions and mechanisms that enhance convenience・Air volume control for Internal air circulation.
|Temperature range (*1)||3℃ to 45℃|
|Temp. control accuracy (*2)||±1.0℃ to 2.0℃|
|Chamber volume||Approx. 90L|
|Outlet inside chamber||1-Socket × 1 (3 A)|
|Connecting hole||On both sides (Each φ36 mm) the Lid is included.|
|Circulating method inside chamber||Forced circulating-type (Manually adjustable on the operation part (*3))|
|Num. of Shelf/Allowable weight||1 level/35kg (Equal weight)|
|Interior/Exterior material||Stainless steel (SUS304) except for some part|
|Safe devices/protections||Overload relay, Temp. sensor error diagnosis, Temp. range over alarm, Memory error alarm, and Compressor High temp. cut circuit|
|Dimensions inside chamber||552(W) x 370(D) x 440(H)mm|
|Int. Dimension||622(W) x 570(D) x 1028(H)mm|
|Power supply (*4)||AC100V/8A /Max.11A (Need a step-down transformer)|
|Standard accessories||Shelf × 1 pc, Pallet × 1 pc, Drip tray × 1 pc, Door key × 1 pc|
|(*1)Defrosting is required once a week when continuous operation at 3℃ in iB-130.|
|(*2)Useable ambient temperature 5℃ to 30℃. The accuracy does not include the fluctuations when switching the solenoid valve is not included.|
|(*3)Strengthen it when there is a lot of condensation and Weaken it when the sample is dry.|
|(*4)When the outlet inside the chamber is used, the power will increase according to the power supply|
|capacity of the device used.|
|•1 set of Shaker NR-1/2/3 can be put on the bottom of the chamber in iB-130. There is a limitation on the shaking speed for use. Described for details in the manual of these products. Any devices that do not generate large vibrations such as a Rotator or Seesaw shaker can be used on the shelf (Each device is verified for operating inside the chamber by TAITEC).|