Constant temperature Chambers (Small size)
In vitro Box iB-130
Code No.： 0077960-000
Most interior and exterior made of stainless steel. High quality Constant temp. chamber Possible for operation of small devices inside chamber.
Applications●Storage of Samples and Reagents
●Cultivation and Immunoprecipitation putting Shaker or Rotator.
High quality In vitro Box unique to TAITECInside and outside are made of stainless steel. It ’s highly durable and easy to clean.
Small size shaker and culture rotator can be entered.Small size shaker(NR-1/2/3 etc.) can be placed at the bottom of iB-130. Only devices with low vibration（RT series etc.）can be placed on the upper stage. There is one outlet (up to 3A) inside.
Functions and mechanisms that enhance convenience・Air volume control for Internal air circulation.
|Temperature range (*1)||3℃ to 45℃|
|Temp. control accuracy (*2)||±1.0℃ to 2.0℃|
|Chamber volume||Approx. 90L|
|Outlet inside chamber||1 x 1-Socket (3A)|
|Connecting hole||On both sides (Each φ36 mm) Lid included.|
|Circulating method inside chamber||Forced circulating-type (Manually adjustable on the operation part in iB-130 (*3))|
|Num. of Shelf/Allowable weight||1 level/35kg (Equal weight)|
|Interior/Exterior material||Stainless steel (SUS304) except for some part|
|Safe devices/protections||Overload relay, Temp. sensor error diagnosis, Temp. range over alarm, Memory error alarm, Compressor High temp. cut circuit|
|Dimensions inside chamber||552(W) x 370(D) x 440(H)mm|
|Int. Dimension||622(W) x 570(D) x 1028(H)mm|
|Power supply (*4)||AC100V/8A|
|Standard accessories||1 x Shelf (iB230: 2 Shelves), 1 x Pallet, 1 x Drip tray, 1 x Door key (iB230: 2 Door keys)|
|(*1) Defrosting required once a week when continuous operation at 3℃ in iB-130.|
|(*2) Useable ambient temperature 5℃ to 30℃. The accuracy not include fluctuation when switching the solenoid valve not included.|
|(*3) Strengthen it when there is a lot of condensation and Weaken it when the sample dried.|
|(*4) When the outlet inside chamber used the power will increase according to the power supply capacity of device used.|
|●1set of Shaker NR-1/2/3 can be put on the bottom of chamber in iB-130 and 1 set of Shaker NR-1/2/3 can be put there in iB-230.There is a limitation on shaking speed for use. Described for detailas in the manual of these products. Any devices that not generate large vibrations such as Rtator or Seesaw shaker can be used on the shelf (Each device verified with operating it inside chamber by us/TAITEC).|