Constant temperature incubator shaker for well-plate
Code No.： 0062288-000
【discontinued】Larger type of shaking incubator which is best for 96-wells Equipped with reverse mode, and temperature could be adjusted even at low temperature.
Applications●Incubating E. Coli and yeast in deep well
●ELISA and panning phage display
●Experiments of digestion of enzyme from mouse tail and dissolving test of hardly soluble materials
FeaturesThis product has been discontinued.
Huge capacity enables temperature adjustment from +4 ºC.Maximizer has freezer system within its large body. It can control temperature +4℃ – +70℃.
Work effectively by original system and mixing speed at 200 – 1500 rpm.Strongly mix micro plate / deep well plate. When incubating E.Coli and yeast by shaking, the same cell growth as with Erlenmeyer’s flask can be obtained in those vials which are not originally suitable for incubation.
Max 24 micro plates or 12 deep wellsShaking rack (option) can be installed up to 4 levels for micro plate (6 plates per level) and 2 levels for deep wells (6 pates per level).
Equipped with reverse shaking mode (available below 1000 rpm)Turning ON this reverse shaking function can turn rotation direction at any timing. Though maximum speed at reverse mode is 1000rpm, this function increase efficiency of shaking and helps dissolving hardly soluble materials.
|Temperature Range||+4 ºC - +70ºC|
|Temp. Control Accuracy||±0.3 ºC - 1.0 ºC(*1)|
|Shaking Method||Horizontal eccentric vibration (with reverse mode)|
|Shaking Speed||200 - 1500rpm (up to 1000 rpm for reverse mode)|
|Applicable Container / Number||Deep well: max 12|
Micro plate: max 24(*2)
|Surroundings temperature||+5 ºC - +35ºC|
|Reverse Shaking Function||ON/OFF, reversing time and speed can be adjusted|
|Heating System||heater (800W)|
|Display (Temp. and speed)||digital display|
|Programming Function||4 pattern, 9 segment, temperature·shaking speed time (for each segment)|
[time setting range: 00h00min - 99h59min], "Wait" function (*3)
|Dimensions||600 x 732 x 643Hmm|
|(*1) Does not include temperature change right after defrost function turn on.|
|(*2) Depends on number of shaking rack (option; see the list below)|
|(*3) "Wait" function is start shaking after reaching the desired temperature.|
|●Shaking rack is option.|